From Glossary of Meteorology
An instrument for determining the dewpoint.
The most widely used systems employ a small polished-metal reflecting surface, cooled and heated electrically using a Peltier-effect device, and a small temperature sensor embedded on the underside of the mirror. An electro-optical system is used to detect the formation of condensate and to provide the input to the servo-control system to regulate the temperature of the mirror. The specular reflectance of the bare mirror decreases with dew layer thickness; a preset threshold defines a specific constant dew layer thickness that is maintained by the optical servo-control. The mirror temperature equals the dewpoint.
Copyright 2022 American Meteorological Society (AMS). For permission to reuse any portion of this work, please contact firstname.lastname@example.org. Any use of material in this work that is determined to be “fair use” under Section 107 of the U.S. Copyright Act (17 U.S. Code § 107) or that satisfies the conditions specified in Section 108 of the U.S.Copyright Act (17 USC § 108) does not require AMS’s permission. Republication, systematic reproduction, posting in electronic form, such as on a website or in a searchable database, or other uses of this material, except as exempted by the above statement, require written permission or a license from AMS. Additional details are provided in the AMS Copyright Policy statement.