The instrument consists of a radiation
sensing element enclosed in a casing that is closed except for a small aperture
through which the direct solar rays enter. Pyrheliometers can be classified on the basis of the sensing elements employed. In one form the sensing element
is a blackened water calorimeter
. The rise in the temperature
of the water gives a measure of the amount of radiant energy
absorbed during the exposure
of the instrument. Another type of sensing element consists of a blackened plate of high heat capacity
. When radiation is allowed to fall on the plate for a period short compared to the thermal
time constant, the temperature rise of the plate is proportional to the intensity of the incoming radiation. A third type of sensing element consists of a pair of plates, one blackened and one reflecting, that are continuously exposed to the incoming radiation. The temperature differential between the plates is proportional to the intensity of the incoming radiation. See Hand (1946) for descriptions of various types of pyrheliometers, for example, silver- disc pyrheliometer, water-flow pyrheliometer
, Eppley pyrheliometer, spectropyrheliometer
, Michaelson actinograph
Hand, I. F. 1946. Pyrheliometers and Pyrheliometric Measurements. U.S. Weather Bureau.
Copyright 2022 American Meteorological Society (AMS). For permission to reuse any portion of this work, please contact firstname.lastname@example.org. Any use of material in this work that is determined to be “fair use” under Section 107 of the U.S. Copyright Act (17 U.S. Code § 107) or that satisfies the conditions specified in Section 108 of the U.S.Copyright Act (17 USC § 108) does not require AMS’s permission. Republication, systematic reproduction, posting in electronic form, such as on a website or in a searchable database, or other uses of this material, except as exempted by the above statement, require written permission or a license from AMS. Additional details are provided in the AMS Copyright Policy statement.