In its basic form it consists of two dissimilar metallic electrical conductors connected in a closed loop. Each junction forms a thermocouple. One thermocouple is maintained at a known temperature
(usually 0°C or a measured temperature) and the other thermocouple is used to measure the unknown temperature. The signal
voltage is a function of the temperature, and the smooth curve can be handled with a simple linear
fit over a moderate temperature range
. Different materials have different curves. Popular thermocouples (and change in voltage per °C) include iron- constantan (50 mv per °C), copper-constantan (38 mv per °C), and various platinum alloys. Thermocouples are also important in home furnaces to detect the pilot light or that the fuel has ignited. A chain of thermocouples, called a thermopile
, can be used as a power supply if a source of heat and cold is available.
Copyright 2022 American Meteorological Society (AMS). For permission to reuse any portion of this work, please contact firstname.lastname@example.org. Any use of material in this work that is determined to be “fair use” under Section 107 of the U.S. Copyright Act (17 U.S. Code § 107) or that satisfies the conditions specified in Section 108 of the U.S.Copyright Act (17 USC § 108) does not require AMS’s permission. Republication, systematic reproduction, posting in electronic form, such as on a website or in a searchable database, or other uses of this material, except as exempted by the above statement, require written permission or a license from AMS. Additional details are provided in the AMS Copyright Policy statement.