It consists of a conventional bulb connected to a capillary in which a constriction is placed so that upon reversal the mercury column
breaks off in a reproducible manner. The mercury
runs down into a smaller bulb at the other end of the capillary, which is graduated to read temperature. A 360° turn in a locally widened portion of the capillary serves as a trap to prevent further addition of mercury if the thermometer
is warmed and the mercury expands past the break-off point. The remote-reading potentialities of reversing thermometers make them particularly suitable for use in measuring temperatures at depths in the sea. In this application, both protected thermometers and unprotected thermometers are used, each of which is provided with an auxiliary thermometer
. They are generally used in pairs in Nansen bottles
. They are usually read to 0.01°C, and after the proper corrections have been applied, their readings are considered reliable to 0.02°C. Details of the correction
procedure are given in Lafond's Tables
Copyright 2022 American Meteorological Society (AMS). For permission to reuse any portion of this work, please contact email@example.com. Any use of material in this work that is determined to be “fair use” under Section 107 of the U.S. Copyright Act (17 U.S. Code § 107) or that satisfies the conditions specified in Section 108 of the U.S.Copyright Act (17 USC § 108) does not require AMS’s permission. Republication, systematic reproduction, posting in electronic form, such as on a website or in a searchable database, or other uses of this material, except as exempted by the above statement, require written permission or a license from AMS. Additional details are provided in the AMS Copyright Policy statement.